RTD Transmitter Calibration Using Fluke 744 – Video Tutorial
This is RTD transmitter calibration video. This video show step by step how to calibrate the RTD transmitter. To calibrate the RTD transmitter the fluke 744 can be used. First Connect the RTD transmitter with the fluke 744. This connection need three wires, but the fluke doesn’t has three connection. So we use the jumper. Here is the tutorial video:
After the RTD transmitter is connected with the fluke, then turn the fluke on. Then press the RTD button to choose RTD. Then input the temperature in Celsius. Then change the temperature into Fahrenheit by pressing the set up menu that shows several parameter that can be set up. After convert the temperature into Fahrenheit, connect the fluke to the communicator.